Conxport Anesthesia Workstation
Anesthesia Workstation with its ergonomic design, integrates a concise and smooth structure and high mobility.
The microcomputer-controlled Ventilator adopts exported sensors and other components. Its scientific and proper design as well as rigorous production process are what demonstrate the inner quality of the Product.
Mechanical ventilation is made secure and visible by the folded respiratory bellows and the security of air supply is guaranteed by pneumatic transmission and measurement system. With multiple modern technologies applied, the Anesthesia Workstation is both outstanding in performance and versatile in functions.
The Ventilator adopts microcomputer control with high-definition 10.4 TFT color LCD display, integrating the functions of respiratory control and displaying the monitoring parameters. It is versatile in function and excellent in performance.
The high-precision Vaporizer is designed with a stable concentration output, automatic compensation for pressure, temperature and flow rate. And it is available to choose two kinds of gas from Halothane, Enflurane, Isoflurane, and Sevoflurane, ensuring a successful anesthesia operation.
High-definition five tube flowmeter (oxygen, nitrous oxide, and air) are designed with stoppers for oxygen and nitrous oxide to control the output oxygen concentration.
Reliable pneumatic transmission system.
Part of Anesthesia Workstation
Main Body : High-strength engineering plastic rack, light, beautiful and corrosion resistant
Scope of application : Adult and Child
Gas source : O2, N2O, Air: 0.28~0.6MPa
Flow meter : O2: 0.05~1.0L/min 1.1~10L/min
N20: 0.05~1.0L/min 1.1~10L/min
Air : 0.05~10L/min
O2, N2O linkage and N2O Stopper
When using nitrous oxide, oxygen concentration>25%;
When the oxygen pressure<0.2Kpa, the flow of nitrous oxide would be cut off.
Flow rate of rapid oxygen supply : 25~75L/min
Low oxygen pressure alarm : There will be sound alarm when the oxygen pressure < 0.2MPa
Vaporizer : It has the function of automatic compensation based on pressure, temperature, and flow rate. The regulation range of evaporator concentration is 0~5 vol%. Among Halothane, Enflurane, Isoflurane and Sevoflurane, two can be chosen for application as required by the customer.
Respiratory Circuit : Working mode: all-close, semi-close, semi-open
Respiratory Bellows : Bellows for adults, tidal volume range : 0~1500 ml
Bellows for children, tidal volume range : 0~300 ml
Anyone can be chosen for application
Part of Ventilator
Display Mode : High-definition 10.4 TFT color LCD screen display
Ventilation Mode : VCV, PCV, SIMV, MANUAL
Ventilation Functions : PEEP, SIGH, IP, IRV, Standby
Ventilation Parameters : Tidal volume 20~1600 ml
Rate 2~99 bpm
SIMV rate 2~20 bpm
Inspiratory trigger pressure : -10~20cmH2O
PEEP : 2~20cmH2O
Pressure Range : 5~60cmH2O
SIGH 1.5 times the inspiratory time (60-120 adjustable)
Inspiratory platform : 0~50%
Pressure control : 3~50cmH2O
Pressure slope : 1~10 gear
Parameters for ventilation monitoring : Expired tidal volume, Inhale tidal volume, minute ventilation volume, IPPV rate, SIMV rate, total respiratory rate, autonomous inspiratory rate, I/E, peak pressure of airway, average pressure, PEEP,inspiratory trigger pressure, Inspiratory platform, oxygen concentration, battery capacity, pressure - time waveform, flow rate - time waveform,flow-volume loop, pressure-volume loop
Security Alarm System
Oxygen Concentration alarm : Upper limit setting range 21%~99%
Low limit setting range 18%~80%
Airway pressure alarm : upper limit setting range 5~60cmH2O
low limit setting range 0~50cmH2O
Per-minute ventilation volume alarmupper limit setting range 1~20L/min
low limit setting range 0~19L/min
Respiratory rate : upper limit setting range 10~99bpm
low limit setting range 0~50bpm
Tidal volume : upper limit setting range 10~1500ml
low limit setting range 0~1400ml
Sustained high-pressure alarm it will give alarm when stress have consistently been higher than 25cm H2O
Battery voltage alarm
Intubation off alarm
Power : AC 220V 50Hz(UPS, Storage battery)
Mechanical Arms : (End-tidal CO2 monitoring)
(Specific interface of waste anesthesia gas)
B No. 4353, Main Road, Near Geeta Gopal Temple,, Ambala Cantt, Haryana, 133001, India